PASSIVE PROBE 60MHZ HIGH VOLTAGE HANTEK P7060
The high voltage passive probe P7060.
These passive high impedance probes are designed and calibrated for use on any oscilloscope that has an input impedance of 1 MΩ and whose input capacity is within the compensation range (refer to the specifications). When connecting the probe, please connect it to the oscilloscope before testing signals. When disconnecting the probe, first disconnect the probe tip from the test signal. In the process of test, make sure that alligator clip can be grounded reliably.
Low frequency response can be matched to the oscilloscope by adjusting the compensation trimmer on the head of the probe.
1. Connect the probe to the oscilloscope and to a 1 KHz square waveform source.
2. Let the oscilloscope display a stable waveform.
3. Carefully adjust the trimmer tool to obtain the flattest tops to the square waves displayed on the oscilloscope.
<div><span style="color: #333333; font-family: 'comic sans ms'; font-size: large;"><br /></span></div>